I recently purchased a Trimble 262 receiver to use with my FMX for trueguide. Yeah have it connected according to the diagram in the manual, 262 into port A. I have configured the receiver with the Ag remote in the FMX and I can get an RTK fix when I’m in Ag Remote. When I press ok and go back to my diagnostic screen it shows a fix but the time starts increasing until it eventually goes to float. If I go back and reconfigure it will fix again but as soon as I exit will do the same thing. Any help is appreciated.